Alaris syringe pumps are manufactured by CareFusion – using non-recommended syringes in Alaris syringe pumps that have a broken spring in the plunger assembly may cause unintended bolus of medication” MHRA (2016).
- If the pump keeps showing ‘check syringe’ alarms, contact CareFusion to discuss how to check the springs housed in the plunger holder.
- Use only the recommended syringes listed in the instructions for use for these pumps.
- Ensure all syringe pumps are regularly maintained and serviced in accordance with the manufacturer’s recommendations.
ReTweet if useful… Risk of uncontrolled IV bolus when using non-recommended syringes http://ctt.ec/FbAE2+ @ivteam #ivteam
Click To Tweet
Thank you to our partners for supporting IVTEAM